Breast pump system with collection container

ABSTRACT

Systems and methods for pumping milk from a breast, wherein the milk is expressed from the breast under suction and milk is expulsed from the pumping mechanism to a collection container under positive pressure.

This application is a Continuation of U.S. Ser. No. 16/050,201, filedJul. 31, 2018, which is a continuation of PCT/US17/17112, filed Feb. 9,2017, which claims priority to Provisional Application Nos. 62/450,528,filed Jan. 25, 2017, 62/442,008, filed Jan. 4, 2017, 62/421,263, filedNov. 12, 2016, and 62/293,480, filed Feb. 10, 2016, the contents of eachof which are incorporated herein by reference in their entirety; and

This application is a Continuation-In-Part of U.S. Ser. No. 16/934,380,filed Jul. 21, 2020, which is a Continuation of U.S. Ser. No.16/667,749, filed Oct. 29, 2019, now U.S. Pat. No. 10,722,624, which isa Continuation of U.S. Ser. No. 15/180,345, filed Jun. 13, 2016, nowU.S. Pat. No. 10,688,229, which is a Continuation of PCT/US2015/041257,filed Jul. 21, 2015, which claims priority from Provisional ApplicationU.S. Ser. No. 62/027,685, filed Jul. 22, 2014, the contents of each ofwhich are incorporated herein by reference in their entirety.

FIELD OF THE DISCLOSURE

The present disclosure generally relates to portable breast pump systemsand methods for collecting milk from a breast of a nursing mother.

BACKGROUND OF THE DISCLOSURE

As more women become aware that breastfeeding is the best source ofnutrition for a baby, and also offers health benefits to the nursingmother, the need is increasing for breast pump solutions that areuser-friendly, quiet, discrete and versatile for use by a nursing motherin various situations. This is particularly true for the working mother,who is away from the home for eight to ten hours or more and needs topump breast milk in order to have it available for her baby, but it isalso a requirement for many other situations where the mother is awayfrom the privacy of the home for an extended period, such as duringshopping, going out to dinner or other activities.

Although a variety of breast pumps are available, a number are awkwardand cumbersome, requiring many parts and assemblies and being difficultto transport. Hand pump varieties that are manually driven are onerousto use and can be inconvenient to use. Some powered breast pumps requirean AC power source to plug into during use. Some systems are batterydriven, but draw down the battery power fairly rapidly as the motorizedpump continuously operates to maintain suction during the milkextraction process. Many of the breast pumps available are clearlyvisible to an observer when the mother is using it, and many also exposethe breast of the mother during use.

There is a continuing need for a small, portable, self-powered, energyefficient, wearable breast pump system that is easy to use, that mimicsnatural nursing, and is discrete by not exposing the breast of the userand nearly unnoticeable when worn.

To ensure that the nursing baby is receiving adequate nutrition, it isuseful to monitor the baby's intake. It would be desirable to provide abreast pump system that easily and accurately monitors the volume ofmilk pumped by the system, to make it convenient for the nursing motherto know how much milk has been extracted by breast pumping. It wouldalso be desirable to track milk volume pumped per session, so that thevolume of milk contained in any particular milk collection container canbe readily known.

There is a continuing need for a breast pump system that is effectiveand convenient to use. The present disclosure addresses these and otherneeds.

SUMMARY OF THE DISCLOSURE

Briefly and in general terms, the present disclosure is directed towardbreast pump systems or methods. The system includes breast contactingstructure and a collection or storage container or assembly, andstructure that delivers milk from a breast to the collection assembly.The method involves pumping milk from a breast and delivering the pumpedmilk into the collection assembly or storage container. In oneparticular aspect, the breast pump system responds in real time tooptimize pumping action for a particular user during a particularpumping session.

According to one aspect of the present disclosure, a system for pumpingmilk from a breast includes one or more of: a skin contact member orflange configured to form a seal with the breast; a conduit in fluidcommunication with and connected to the skin contact member; a drivingmechanism configured to establish a vacuum profile within the conduit;an external shell; a milk collection container; and a non-transitorycomputer readable medium having stored thereon instructions executableby a computing device to cause the computing devices to performfunctions associated with and directed by the instructions; wherein theexternal shell comprises a compartment; wherein the skin contact member,the conduit and the driving mechanism are received in the compartment ofthe external shell; wherein the milk collection container ispositionable within the shell; and wherein the system is shaped andconfigured to be contoured to the breast of a user.

In one or more embodiments, the system can include one or more ofstructure configured to address fluid ingress, pinch protectionstructure, a flex-tube structured to accomplish efficient andpredictable pumping of fluid and the creation of desirable pressureprofiles, and cooperating structure for fluid collection secureattachment and removal.

In various embodiments, the storage container can be specificallyconfigured to prevent kinking and for durability and handling. Thestorage container can be designed to hold, accept or retain milk orother fluids. A flow feature can be incorporated into the storagecontainer in the form of a scallop structure, valves and materials canbe chosen to facilitate removing air or gases, tabs and wings can beprovided for handling, and structure adapted for the removal of milkfrom a collection assembly.

In various of the disclosed embodiments, the system defines a breastprofile. The natural breast profile is contemplated to fit comfortablyand conveniently into a bra of a user and to present a natural look. Assuch, the profile is characterized by having a non-circular base.Moreover, like natural breasts, the profile of the device or system iscontemplated to define one or more asymmetric curves and off-centerinertial centers. In one aspect, the system defines a breast enhancementsystem for enlarging the appearance of the user's breast.

In at least one embodiment, the system functions by operating a controlsystem that tracks internal pressure of the system against a knownwaveform. In this regard, the waveform can be a vacuum waveformindicative of pressures applied to a breast, and can define a sine wavefluctuating between about 60 mmHg of vacuum to a vacuum from about 120mmHg to about 250 mmHg, or other desirable or useful waveform.

In one or more embodiments, the system includes a controller thataccomplishes real time pressure control inside the system.

In one or more embodiments, the system includes a controller providingautomated compliance sensing and response.

In one or more embodiments, the system includes a non-contact pressuresensing arrangement that does not touch the skin or the milk inside thetube while accurately determining internal pressure of the tube.

In one or more embodiments, the system includes one or more controllersthat automatically detects one or more of letdown, overfill and flow.

In one or more embodiments, the system is disabled when the flange isnot placed in an operating position.

In one or more embodiments, the system can be adapted to visualize auser's data and trends as it relates to volume (from each breast andtotal), and number of sessions on several dimensions (per day, per weekand per month). Data and analytics can also be provided on pumpingsession.

In at least one embodiment, the flange or skin contact member, theconduit, the driving mechanism, the external shell and the milkcollection container are all contained within a cup of a brassiere. Inother embodiments, the container need not be contained within thehousing, and the pump need not be in a cup of a brassiere, but can beunsupported or supported by itself or by other clothing or a nursingtank top or a band surrounding the user's body.

In at least one embodiment, the system is battery powered, the systemcomprising a battery, wherein the battery is received in the compartmentof the external shell.

In at least one embodiment, the milk collection container comprises aone-way valve that permits milk inflow into the milk collectioncontainer but prevents milk backflow from the milk collection containerto the conduit. In one embodiment, the collection container or containerassembly includes an extra part, valve or fitment that is attachedthereto and facilitates creating a seal with the container to establisha closed system. In one embodiment, the milk container can include aone-way valve that cannot be removed without destroying milk containeror valve function. The valve can assume a myriad of shapes and kindsincluding an umbrella valve, a duckbill valve, a ball valve or othervalve. Moreover, in one or more embodiments, the container can beflexible or rigid, or disposable or reusable.

According to another aspect of the present disclosure, a system forpumping milk from a breast includes one or more of: a flange or skincontact member configured to form a seal with the breast; a conduit influid communication with and connected to the skin contact member; adriving mechanism configured to establish a vacuum profile within theconduit by cyclically compressing and allowing decompression of aportion of the conduit; and an external shell containing the conduit andthe driving mechanism and supporting the skin contact member.

In at least one embodiment, the system further includes a milkcollection container, wherein the milk collection container is in fluidcommunication with the conduit.

In at least one embodiment, the skin contact member includes: a breastcontact portion configured and dimensioned to fit over and form a sealwith a portion of the breast; and a nipple receiving portion extendingfrom the breast contact portion.

According to another aspect of the present disclosure, a method ofoperating a system for pumping milk includes one or more of: providingthe system comprising a skin contact member configured to form a sealwith the breast, a conduit in fluid communication with and connected tothe skin contact member; a driving mechanism including a compressionmember configured to compress and allow decompression of the conduit inresponse to inward and outward movements of the compression member, asensor, and a controller configured to control operation of the drivingmechanism; sealing the skin contact member to the breast; operating thedriving mechanism to generate predetermined pressure cycles within theconduit; monitoring by the controller of at least one of position andspeed of movement of the compression member relative to the conduit;measuring or calculating pressure within the conduit; maintaining ormodifying motion of the compression member as needed, based uponfeedback from the calculated pressure and at least one of force,position and speed of movement of the compression member, to ensure thatthe predetermined pressure cycles continue to be generated.

In at least one embodiment, the predetermined pressure cycles compriseextraction pressure cycles, and the controller increases a strokedistance of the compression member relative to an amount of milkentering the conduit, to maintain predetermined pressures during theextraction pressure cycles.

In at least one embodiment, the predetermined pressure cycles compriselatch cycles, wherein upon determination that milk has entered theconduit or after a predetermined period of time, the controller operatesthe compression member to achieve predetermined extraction pressurecycles, wherein the predetermined extraction cycles differ from thepredetermined latch cycles by at least one of maximum suction level,cycle frequency or waveform shape. Moreover, in one or more embodiments,the system includes structure or functions to recognize when a user isdone pumping, or includes structure or functions such that when there isa loss of vacuum recognition which allows the user to easily end apumping session by simply pausing and pulling the device off of thebreast. Additionally, in one or more embodiments, the system can includean auto-purge function or an accelerometer functioning as gesturerecognition so that the device can interpret what the user is attemptingto accomplish.

According to another aspect of the present disclosure, a system forpumping milk includes one or more of: a flange or skin contact memberconfigured to form a seal with a breast; a conduit in fluidcommunication with and connected to the skin contact member; a drivingmechanism including a compression member configured to compress andallow decompression of the conduit in response to inward and outwardmovements of the compression member; a sensor; and a controllerconfigured to control operation of the driving mechanism; wherein uponsealing the skin contact member to the breast, the controller operatesthe driving mechanism to generate predetermined pressure cycles withinthe conduit, monitors at least one of position and speed of movement ofthe compression member relative to the conduit, measures or calculatespressure within the conduit based upon signals received from the sensor,and maintains or modifies motion of the compression member as needed,based upon feedback from the calculated pressure and at least one offorce, position and speed of movement of the compression member, toensure that the predetermined pressure cycles continue to be generated.

These and other features of the disclosure will become apparent to thosepersons skilled in the art upon reading the details of the systems andmethods as more fully described below.

BRIEF DESCRIPTION OF THE DRAWINGS

FIG. 1A shows a perspective view of a breast pump system according to anembodiment of the present disclosure.

FIG. 1B is a rear view, depicting the flange of the pump system of FIG.1A.

FIG. 2 shows a front view of the system of FIG. 1 with the shellremoved.

FIG. 3 depicts a back view of the system of FIG. 1 with the flangeremoved.

FIG. 4 is a cross-sectional side view of the system of FIG. 1 .

FIG. 5 is an inside view of the system of FIG. 1 , depicting the flexconduit of the pump assembly.

FIG. 6A is an exploded view of the system of FIG. 1 , depictingmechanical components of the system.

FIGS. 6B-D depict views of an alternative approach to housing structure.

FIGS. 6E-G depict views of yet another alternative approach to housingstructure.

FIG. 6H is a perspective view, depicting a first approach to a systemincluding a removable battery structure.

FIG. 6I is a perspective view, depicting a second approach to a systemincluding a removable battery structure.

FIG. 6J is a perspective view, depicting a third approach to a systemincluding a removable battery structure.

FIG. 7A is a schematic representation, depicting operational componentsof the system.

FIG. 7B is a graphical representation, depicting motor position andvacuum versus time.

FIG. 7C is a graphical representation, depicting motor position versusvolume.

FIG. 8 is a top view, depicting one embodiment of a storage collectionassembly of the present disclosure.

FIG. 9 is an enlarged view, depicting a neck and valve of the storagecollection assembly of FIG. 8 .

FIG. 10 is an enlarged view, depicting a valve assembly of the storagecollection assembly.

FIG. 11A is a perspective view, depicting a storage collection assemblyconnected to the system.

FIG. 11B is a perspective view, depicting a first step in installing acollection assembly.

FIG. 11C is a perspective view, depicting a second step in installing acollection assembly.

FIG. 11D is a top view, depicting a third installation step.

FIG. 11E shows yet another collection assembly installation step.

FIG. 12 is a cross-sectional view, depicting a portion of the system.

FIG. 13 is a perspective view, depicting a door assembly of the system.

FIG. 14 is a cross-sectional view, depicting details of the doorassembly.

FIG. 15 shows details of the door assembly.

FIG. 16 is an enlarged view, depicting structure of a pinch protectionassembly.

FIG. 17 is an enlarged view, depicting other structure of a pinchprotection assembly.

FIG. 18 is a perspective view, depicting a flex circuit of the system.

FIG. 19 is a top view, depicting a user interface assembly.

FIG. 20 is a bottom view, depicting further details of the userinterface assembly.

FIG. 21 shows power access support structure of the system.

FIGS. 22-28 depict various aspects of a remote user interface system.

FIGS. 29-41 depict various further aspects of a remote user interfacesystem.

DETAILED DESCRIPTION OF THE DISCLOSURE

Before the present systems and methods are described, it is to beunderstood that this disclosure is not limited to particular embodimentsdescribed, as such may, of course, vary. It is also to be understoodthat the terminology used herein is for the purpose of describingparticular embodiments only, and is not intended to be limiting, sincethe scope of the present disclosure will be limited only by the appendedclaims.

Where a range of values is provided, it is understood that eachintervening value, to the tenth of the unit of the lower limit unlessthe context clearly dictates otherwise, between the upper and lowerlimits of that range is also specifically disclosed. Each smaller rangebetween any stated value or intervening value in a stated range and anyother stated or intervening value in that stated range is encompassedwithin the disclosure. The upper and lower limits of these smallerranges may independently be included or excluded in the range, and eachrange where either, neither or both limits are included in the smallerranges is also encompassed within the disclosure, subject to anyspecifically excluded limit in the stated range. Where the stated rangeincludes one or both of the limits, ranges excluding either or both ofthose included limits are also included in the disclosure.

Unless defined otherwise, all technical and scientific terms used hereinhave the same meaning as commonly understood by one of ordinary skill inthe art to which this disclosure belongs. Although any methods andmaterials similar or equivalent to those described herein can be used inthe practice or testing of the present disclosure, the preferred methodsand materials are now described. All publications mentioned herein areincorporated herein by reference to disclose and describe the methodsand/or materials in connection with which the publications are cited.

It must be noted that as used herein and in the appended claims, thesingular forms “a”, “an”, and “the” include plural referents unless thecontext clearly dictates otherwise. Thus, for example, reference to “asensor” includes a plurality of such sensors and reference to “the pump”includes reference to one or more pumps and equivalents thereof known tothose skilled in the art, and so forth.

The publications discussed herein are provided solely for theirdisclosure prior to the filing date of the present application. Thedates of publication provided may be different from the actualpublication dates which may need to be independently confirmed.

Various details of the present system can be found in PCT ApplicationNos. PCT/US15/41257, PCT/US15/41271, PCT/US15/41277, and PCT/US15/41285each filed Jul. 21, 2015, and PCT/US15/50340 filed Sep. 16, 2015, eachof which are hereby incorporated herein, in their entireties, byreference thereto.

FIGS. 1A-B are perspective and back views of a breast pump system 10according to an embodiment of the present disclosure. The breast pumpsystem 10 can include one or more of the below introduced or describedfeatures or functions, or a combination thereof. The housing or outershell 12 of system 10 can be shaped and configured to be contoured tothe breast of a user and to thus provide a more natural appearance whenunder the clothing of the user. As can be appreciated from the figures,the system can define a natural breast profile. The natural breastprofile is contemplated to fit comfortably and conveniently into a braof a user and to present a natural look. As such, the profile ischaracterized by having a non-circular base unlike that embodied in agenerally dome-shaped configuration. Extending from the base are curvedsurfaces having asymmetric patterns. Moreover, like natural breasts, theprofile of the device or system is contemplated to define one or moreasymmetric curves and off-center inertial centers. Various naturalbreast shapes can be provided to choose from to the tastes and needs ofa user. An opposite side of the pump system 10 is configured with aflange 14 which is sized and shaped to engage a breast of a user. Theflange 14 is contoured to comfortably fit against a wide range of user'sbodies and to provide structure for sealingly engaging with breasttissue. In one particular embodiment, the flange 14 can form generallyrigid structure, and alternatively or additionally unlike a standardflange can lack sharp edges or a lip portion against which breast tissuemight be engaged during use. In this regard, the flange includessurfaces that extend outwardly from a nipple receiving portion of theflange to engage breast tissue, thus providing extra surface area forcomfortably contacting tissue. Various approaches are contemplated tothe flange with respect to a user's nipple. One approach involvesaligning a horizontal line formed within the flange structure a bithigher than center with the rationale that the perspective of the motheris from above. This perspective allows the user to better align thebreast with the horizontal line to better center the nipple in theactual center of a nipple receiving portion of the flange, thusoffsetting any tendency to aim/line up low associated with centeredlines because the user's perspective is from above the line and alsobecause the device is pivoted into place in certain instances.

FIG. 2 is a front view of the system 10 of FIG. 1 , with the housing orouter shell 12 having been removed and made transparent to showcomponents otherwise covered by the housing 12. In particular, with thehousing 12 removed, various electronic components can be identified. Thesystem controller is embodied in a circuit board 15 that is incommunication with a flex-circuit 16, each cooperating to connect to andcontrol various electro-mechanical components of the system 10. Acontrol panel 17 is in electronic communication with the controller viathe flex-circuit 16 and provides the user with the ability to power thesystem on and off as well as to alter functioning. One or more motors44, 46 are further provided and controlled electronically by the systemto effect manipulation of actuators (described below) operating on aconduit or flex-tube 32 (See FIGS. 4 and 5 ). A battery 48 is includedto provide a rechargeable power source and is configured to be pluggedinto a power source for charging. Further, there is provided a load cellassembly 54 that is configured to provide a pressure sensing function asdescribed below. It is contemplated that at least in one embodiment, theconduit or flex-tube 32 is oriented to run from inferior to superiorrelative to the nipple of a breast when the user is upright.

FIG. 3 shows an opposite side of the system 10 with the flange 14removed to illustrate more details of the pumping function. The conduitor flex-tube 32 (See FIGS. 4-6) includes generally spherically shapedconnectors 33 that are sized and shaped to be removably received inrecesses 34 formed in a pump chassis 35. The connectors 33 are designedto automatically engage with moving motor paddles without the user beingaware or having to make adjustments, or assemble parts. The pump chassis35 functions to support the electronic and electro-mechanical structuresof the system 10 (See also FIG. 2 ). It also provides spacing for apinching actuator 36 that is configured to be advanced and retractedtoward and away from the conduit or flex-tube 32 as described furtherbelow. Other pumping action is accomplished through the engagement ofthe conduit or flex-tube 32 with recesses 34 by a compression andexpansion member 38 (See FIG. 7A). Embossed engraving is furtherprovided within a well formed in the chassis 35, the engraving providingproduct and other information relevant to the breast pump. In this way,certain adhesive labels need not be applied to the breast pumpstructure.

In general, real-time pressure control can be managed by a controller ofthe system 10. The controller tracks pressure and moves a pump motoreither in or out to influence the pressure in the direction of itschoosing. By way of oscillating motion of the motor, the pump can beconfigured to pull on the connectors 33 of the conduit or flex-tube 32structure to increase its volume. If there is vacuum in the system 10that vacuum can be increased as the volume of the tube increases.Pushing in the tube decreases its volume. This in turn causes the vacuumlevel to decrease in the tube, and can cause a relative positivepressure if vacuum decreases enough. The pump controller applies theseprinciples, sensing the current pressure and then nudging a compressionmember or paddle of the motor assembly in a direction required togenerate a pressure target. By doing this repeatedly in real time, thesystem can create a controlled vacuum waveform that matches waveformsdesired to be applied to a user's nipple.

The pump can slowly pull the compression member or paddle out until ithits a pre-determined target. Should the paddle be moved to the end ofits range without being able to generate a desired vacuum, the systemwill be purged to generate more vacuum potential. The purge functions topush material out of the system to create a strong vacuum potential. Itaccomplishes this by first closing a pinch on the conduit or flex-tubeor closing off the flex-tube with a flap, dam, etc., then evacuating theflex-tube, for example, by pushing closed the paddle, which forcesvolume out of the flex-tube and any fluid or air that was inside thatvolume is also ejected through the one-way valve and into the collectionreceptacle. When the paddle retracts again, it can then generate muchhigher vacuum as contents of the tube had been previously purged. Once ahigher vacuum can be generated, the system can open the pinch valve sothat the desired vacuum profile can be applied to a breast and desiredpressure waveform can be produced.

When the system is filled with air, it is very compliant such that alarge change in motor positioning makes only a small change in vacuum.When the system is filled with fluid on the other hand, a small changein motor positioning makes a big change in vacuum. In one particularapproach, an encoder including a plurality of spaced magnets isassociated with the motor. The magnets can be placed along a peripheryof a generally disc shaped encoder with the magnets oriented parallel tothe axis of rotation of the encoder. One or more hall effect sensors canbe configured on or surface mounted to the circuit board 15 andpositioned to read the motion and position of the magnets. In this way,the position of the motor can be determined and monitored. Thus, achallenge can be to configure the system so that it is stable when thesystem is responsive, and effective when it is not as responsive. Onecontemplated approach is to tune the controller for a relatively rigidsystem and to input unit-less quantities that move the motor in requireddirections where the amplitude of which is modified depending on theoutput of the system. Accordingly, a cascade controller can be createdto grow an input wave if system output is smaller than desired to hitpressure targets and can be shrunk if the system output is larger thanrequired. This can be accomplished in real time by observing outputverses input. In this way, the controller can be continuously adjustingtarget waveforms. Top half and bottom half waveforms can haveindependent control which facilitates centering waveforms in aneffective manner, and results in a system that is both very accurate andquick to adjust.

The system can further be provided with automated letdown detection. Thepump can sense when it is full of fluid and responds accordingly byswitching between pumping and letdown when fluid has begun to flow. Inone approach an algorithm incorporated into the system can operate tolook at the ratio of maximum and minimum of a target wave in the pumpand compare that against the output of the pump. The result is aunit-less but very reliable sensing of system compliance. This can betuned to trigger an internal event when the compliance crosses someknown values that represent when the system is full of fluid. Any othermeasurement of compliance can be used in an equivalent way.

In another approach to letdown detection, it is noted that pushing atube of air does not generate the same forces as pushing a tube offluid. Tracking the force generated during a purge can also give astrong indication of when the system is full of fluid. An event can begenerated to track this such that when the force of a purge crosses someknown threshold the system can be said to be full of fluid rather thanair. This approach may involve less tracking of data and less tuningthat is subject to change with pump design or breast tissue. In yetanother approach, letdown detection can be based upon tracking flow.That is, when flow begins, letdown must have occurred and when a smallvolume of flow has been collected the system can switch to pumping.Further, letdown can be tracked by looking at the relative rate ofchange of vacuum measured to motor position. Note that this relativerate of change is a measurement of compliance. As this ratio goes up inmagnitude, it can be concluded that the system is filling with fluid.

FIG. 4 illustrates a cross-section of components of a system 10according to an embodiment of the present disclosure. Flex-tube orconduit 32 (isolated in FIG. 5 ) includes a large conduit portion 32Lthat is relatively larger in cross-sectional inside area than thecross-sectional inside area of small conduit portion 32S. The largeconduit portion 32L terminates with an opening sized for cleaning and isgenerally sized to accept a small finger tip. Although both portions 32Sand 32L are shown as tubular portions, the present disclosure is notlimited to such, as one or both portions could be shaped otherwise. Whentubular, the cross-sections may be oval, square, other polyhedral shape,non-symmetrical, or non-geometric shape. Further, the flex-tube 32 caninclude an enlarged bulbous portion 32B configured near a terminal endof the large conduit portion 32L that is provided to help accommodatesystem hysteresis.

FIG. 6A depicts an exploded view of structural and mechanical componentsof the system 10. Configured between the housing 12 and flange 14 is thechassis 35. Notably, the chassis can be configured to snap intoengagement with the housing 12. Moreover, in a preferred embodiment, thechassis 35 supports directly or indirectly all of the pump components.In particular, a PCB controller mount 62 is supported by the chassis 35and is configured to be connected to and support the circuit board 15(See also FIG. 2 ). A battery bracket 64 is also supported by thechassis 35 and is sized and shaped to receive a rechargeable battery 48assembly that powers the system 10. A cover jack 65 is further includedto provide access to the battery assembly and for accepting a power cordconnector (not shown). Motor mounting 66 and motor receiver structure 67is also supported by the chassis 35 and are configured to receive andsupport the system motor which is powered by the battery and whichfunctions to move actuators operating on the conduit or flex-tube 32.Also supported by the chassis 35 are an actuator bracket 69, and a loadcell bracket 70 and load cell receiver 71. Moreover, user interfacepanel can include a button membrane 72 and a button membrane housing 73each supported on the housing 12 and placed in engagement with theflex-circuit 16 that provides the user with system control.

In order to connect the conduit or flex-tube assembly 32 to the system10, there are provided a flex-tube ring 80 and a flex-tube collar 82.The flex-tube collar 82 is sized and shaped to be received into slots 84on the flange. A fluid container fitment 86 (shown in isolation from thecontainer) is sized and shaped to be received into the flex-tube collar82. A door assembly 90 is attached to the flange 14 and configured toswing open and closed to both provide access to an interior of thesystem 10 as well as to support a robust connection between the fitment86 and flex-tube collar 82. Accordingly, it is contemplated that in atleast one embodiment, the collection or container assembly is supportedand maintained in attachment by friction around a shaft of the conduitto the collection or container assembly, and partially by the doorassembly 90 which can enclose and hold the collection or containerassembly in place. In alternative embodiments, the breast pump assemblycan omit a door assembly entirely. Thus, the flange itself can includestructure for retaining the container assembly in place. Moreover, thedoor assembly or other structure that replaces the door assembly can betransparent so that a direct view to the container assembly is provided.

In alternative embodiments, the housing 14 is defined by an irregularshape that includes contours that track or mimic the internal componentsand structure of the pump system. In one particular approach as shown inFIGS. 6B-D, an outer surface of the housing 14 is characterized by anirregular shaped indentation 91, providing the outer surface with anirregular shape. Various differently shaped indentations 91 can beemployed (See also FIGS. 6E-G). Various configurations of a separatebreast cup skin or interface structure 98 is sized and shaped to fitover the housing 14 and indentation 91 to form a desired shape such asthe breast shape depicted in the figures. It is noted that the breastpump system can work with or without the breast cup skin or interfacestructure. Alignment and attachment structure or holes can be furtherprovided to facilitate the mating of the interface structure 91 with thehousing 14 and the interface structure can assume a myriad of colors,textures and durometer to enhance or change tackiness, softness forsecurity and outer feel in the bra. Various other breast and othershaped can also be provided.

In yet further combined or separate embodiments (See FIGS. 6H-J), thehousing 14 can be adapted or configured to additionally or alternativelyaccommodate a replaceable battery. Here, the housing 14 includes variousother shaped indentations 91 sized and shaped to accommodate thebattery. In this approach, the battery includes its own attachablehousing 99 that mates with the housing 14 indentation 91, the housing 14covering other pumping structures. In one approach, mating featuresinclude flat, right-angled structures, and alignment and attachmentholes and structures are further provided.

As shown schematically in FIG. 7 , latching, pumping and extractionforces can be established by two compression members 36, 38 that areactively driven by motor drivers 44 and 46 respectively. Although morethan two compression members could be used and one or more than twodrivers could be used, the currently preferred embodiment uses twocompression members respectively driven by two drivers as shown. Asystem controller or system software and/or firmware controls the actionof the drivers in real time, responsive to pre-determined latching andproduction targets or schemes as detected by the pressure sensor or loadcell assembly. The firmware can be written so that such targets can beapproached at various speeds, sometimes relatively quickly and othertimes more slowly or gently to thereby provide multiple stimulation andexpression levels. Thus, for example, latch can be achieved takingalternatively more gradual or quicker approaches, and there can becontrols determining the level at which latch is achieved. Variouslevels of suction can be present during expression as well. Tubingportions 32S and 32L can be closed off, or substantially closed off bycompression members 36 and 38, respectively. Moreover, such activepumping members can be configured to engage upon a tubing channelgenerally perpendicularly to the net flow of fluid or milk within thechannel. Also, a pinch region of the tubing channel can be configured toopen through passive recoil located next to a compression region of thetubing channel which opens through an assistive active support. Uponpowering up the system 10 the compression member 36 opens and thecompression member 38 begins to withdraw away and through its connectionto structure such as the ball connector of the conduit or flex-tube 32thereby gradually increases the suction level within tubing 32. When apredetermined maximum suction level is achieved (as confirmed bypressure readings taken from a pressure sensor, described below), thecompression member 38 ceases its travel in the current direction, andeither maintains that position for a predetermined period of time (ormoves slightly in the same direction to compensate for decreasingsuction as milk enters the system) when the operating mode of the system10 has a predetermined time to maintain maximum suction, or reversesdirection and compresses the tube 32L until the latch suction level isachieved. If the maximum suction level has not yet been achieved by thetime that the compression member can be fully retracted away on thefirst stroke, then the compression member 36 again compresses the tube32S to seal off the current vacuum level in the environment of thebreast, and the compression member 38 fully compresses the tube portion32L to squeeze more air out of the system. Then the compression member36 reopens to fully open tube portion 32S and compression member carriesout another stroke, again moving away to generate a greater suctionlevel. This cycling continues until the maximum suction level isachieved. It is noted that it is possible in some cases to achieve themaximum suction level on the first stroke, whereas in other cases,multiple strokes may be required.

Upon achieving the maximum suction, the system may be designed andprogrammed so that the compression member 38 does not travel to itsfullest possible extent in either direction to achieve the maximum andlatch suction levels, so as to allow some reserve suction and pressureproducing capability. When the maximum suction level has been achieved,and the pumping profile can return to latch vacuum, the compressionmember 38 advances compressing tubing portion 32L, thereby raising thevacuum in the tubing 32. Upon achievement of the latch suction vacuum,compression member 36 closes off the tubing 32S again to ensure that thelatch vacuum is maintained against the breast, so that sufficientsuction is maintained. At this stage, the compression member 38 againbegins moving away to increase the suction level back to maximumsuction, and compression member 36 opens to allow tube 32S to open andthe breast 2 to be exposed to the maximum suction. Alternatively, thesystem may be programmed so that the compression member 38 cyclesbetween maximum and latch suction levels without the compression member36 closing during a point in each cycle, with the compression member 36closing when the latch vacuum is exceeded.

Upon commencing milk extraction, the compression member 36 andcompression member 38 can function in the same manner as in latching,but in a manner that follows an extraction waveform determined by theselected extraction pumping determined in real time by system controlswhich are responsive to the load cell assembly or pressure sensingassembly. At this stage, any sounds created by the pumping action of thesystem are decreased as milk or fluid flows through the pump mechanism.During the compression stroke of compression member 38, compressionmember 36 closes when the latch pressure/suction level is achieved.Continued compression by the compression member 38 increases thepressure in the tubing 32 downstream of the compression member 36 toestablish a positive pressure to drive the contents (milk) of tubeportion 32L out of the tube portion 32L through smaller tubing portion32S2 downstream of 32L and out through a one-way valve. The positivepressure attained is sufficient to open the one-way valve for deliveryof the milk out of the tubing 32 and into a milk collection container.In one embodiment, the positive pressure is in the range of 20 mm Hg to40 mm Hg, typically about 25 mm Hg. Upon reversing the motion ofcompression member 38, compression member 36 opens when the suctionlevel returns to the latch suction level and compression member 38continues to open to increase the suction level to the maximum suctionlevel.

The present disclosure can establish a latch vacuum to cause the flangeor skin contact member/breast 14 to seal to the breast. The latch vacuumestablished by the system is currently about 60 mmHg, but can be anyvalue in a range of from about 20 mmHg to about 100 mmHg. Once thesystem 10 has been latched to the breast via skin contact member 14, thesystem then cycles between the latch vacuum and a target (also referredto as “peak” or “maximum”) suction level. Due to the fact that thesystem 10 does not cycle down to 0 mmHg, but maintains suction appliedto the breast, with the minimum end of the suction cycle being the latchsuction level (e.g., about 60 mm Hg), the nipple does not contract asmuch as it would with use of a prior art breast pump system. It has beenobserved that the nipple draws into the skin attachment member 10 withthe initial latch achievement in an analogous fashion as the formationof a teat during breastfeeding. Once the vacuum cycles between the latchand target vacuum levels, there is significantly less motion of thenipple back and forth with the vacuum changes, as compared to whatoccurs with use of prior art systems. The nipple motion (distancebetween fully extended and fully retracted) during use of the presentsystem is typically less than about 2 mm, and in some cases less thanabout 1 mm. Accordingly, the system provides latching that is not onlymore like natural nursing, but the reduced nipple motion is also morelike natural nursing as evidenced by scientific literature. In oneparticular approach, the system can employ ultrasound to observe nipplemotion during pumping to ensure that desired nipple motion is achieved.

This greatly reduced motion of the nipple during cycling results fromestablishment of the latch at latch vacuum level, and then limiting therange of vacuum swing between latch vacuum (suction) and peak vacuum(suction). Typically the difference in vacuum between latch vacuum andpeak vacuum is less than 200 mmHg, more typically less than 150 mmHg. Inone example, the latch vacuum was 50 mmHg and the peak vacuum was 200mmHg, resulting in a vacuum difference of 150 mmHg.

Limiting the nipple motion as described with use of the present systemoffers several benefits to the user. One benefit is that there is lessfriction on the side of the nipple against the flange wall, therebygreatly reducing the risk of irritation, skin damage, pain, swelling,etc. As a result, the present system is significantly more comfortableto use by a nursing mother, and this benefit is increasingly noticeableover repeated uses. By maintaining at least a latch suction level at alltimes, the present system provides a more secure and persistent seal tothe breast and significantly reduces the potential for leaks of airand/or milk. Because the nipple moves significantly less, this providesa more “natural” feel to the user that more closely simulates the feelof a nursing baby. Because the nipple travels less, this allows for theskin attachment member/flange 14 to be designed as a lower profilecomponent, as its length can be shorter since it does not need toaccommodate the greater length in nipple movement experienced by priorart systems. This allows the overall amount of protrusion of the system10 from the breast to less than that in the prior art, as the overalllength of the system is reduced by the reduction in length of the skincontact member/flange 14. Thus, the distance from the tip of nipple toexposed end of the housing the system is reduced.

The breast contact portion can be symmetrical about the nipple receivingportion although, alternatively, the nipple receiving portion could beoffset. The skin contact member 14 is designed to reduce the internalvolume of the nipple receiving portion, which is enabled by thesignificantly reduced amount of motion experienced by the nipple duringa milk extraction process using a system 10 including skin contactmember 14, according to the present disclosure. The nipple receivingportion of the skin contact member 14 is contoured to more closely matchthe natural shape of the nipple, thereby eliminating or significantlyreducing dead space that exists around the nipple in prior art systems.The nipple receiving portion can be cylindrical in the portion adjoiningthe breast contact portion, and then can taper conically. This designallows for receiving a portion of the areola into the nipple receivingportion while also limiting dead space. The diameters of allcross-sections of the nipple receiving portion are contemplated to belarge enough to allow nipple dilation. The length of the nipplereceiving portion can be about 23 mm and the length may vary within arange of about 22 mm to about 29 mm. The length of the nipple receivingportion is sufficient to allow engorgement of the nipple under vacuum,without the distal tip of the nipple contacting the proximal end of thenipple receiving portion. In an alternative approach, the nipplereceiving portion can be sized and/or shaped to mimic the anatomy of achild that is nursing. In this regard, rather than being generallycylindrical, the nipple receiving portion define more of a natural mouthshape or a generally rectangular sleeve with rounded corners and curvedsurfaces. The teat of the breast is thus formed into a more naturalnursing shape by the natural shaped nipple receiving portion.

The internal contour 120 of the flange 14 is designed for use with thepresent system 10 and to maximize comfort of the user. The internalangles and generally flat portions also facilitate the ability torestrict portions of the breast from moving forward too much into thenipple receiving portion. The wider angle helps to prevent the breasttissue from being funneled into the nipple receiving portion, so thatless breast tissue is received in the nipple receiving portion, makinguse of the flange 14 more comfortable than flanges of the prior art andproviding space for nipple engorgement. By providing the wider angles,this also allows the overall system to be effectively shortened andallows the system to lie flatter against the breast to improve bothcomfort and appearance.

In one embodiment, the total system volume is about 24.0 cc. The totalvolume is calculated as the space in the nipple receiving portion (thatis not occupied by the nipple) and tube portions 32S, 32L and 32S2 up tothe milk collection or container assembly. In the embodiment with totalsystem volume of about 24.0 cc, the active pump volume, i.e., the volumedisplacement achievable by compressing tube portion 32L from fullyuncompressed to the limit of compression by compression member 38 isabout 3.4 cc. When there is only air in the tubing 32 of the system 10,pressure swing by moving the compression member 38 inwardly against thetubing portion 32L and outwardly away from the tubing portion islimited, due to the compressibility of the air. In this embodiment, withthe system under vacuum of −60 mmHg, a full stroke of the compressionmember (from compressed to fully uncompressed tube portion 32L)increases the vacuum to −160 mmHg. The ratio of pumping volume to totalsystem volume can be important with regard to power and size of thepumping system. In this embodiment, the tube portion 32L was made ofsilicone. It has been recognized that reduced motion of the compressionmembers when pumping allows for more quiet action of the pump motor, anda more quiet system overall. Further, the present system employs themilk expressed as the medium for system hydraulics, and this medium isin direct contact with the user's breast against which a vacuum isdrawn. Thus, the system can employ air suction against the breast forinitial latching and pumping and then converts to utilize expressedbreast milk for pumping action or power.

During let down operation, the system 10 operates to effect let down ofthe milk in the breast, prior to extraction, with a maximum suctiontarget of up to 120 mmHg (typically, about 100 mmHg (−100 mmHgpressure)) to establish let down. The goal of letdown (or non-nutritivesuction) is to stimulate the breast to express milk. The relativelyshallow (small vacuum change range) and relatively fast frequency of thepumping during this phase are meant to mimic the initial suckling actionof a child at the breast. This is because during let down phase, thesuction pressure is not allowed to exceed the maximum let down suctionof 110 mmHg or 120 mmHg, or whatever the maximum let down suction is setat. Therefore, as the compression member 38 is drawn in a direction awayfrom the tube portion 32L, the system 10 is designed to reach −100 mmHg(a suction pressure of 100 mmHg) (or −120 mmHg, or whatever the maximumlet down suction is designed to be), by the time that the compressionmember 38 has reached a position in which tube 32L is mostlyuncompressed.

During let down (non-nutritive) the system software and/or firmwarecommunicates instructions to system motors based upon readings taken andcommunicated from the pressure sensing assembly so that the system isconfigured to operate between −60 mmHg and −100 mmHg in one example. Inthis example, the compression member 38 can compress the tubing portion32L nearly fully and then be moved away from the tubing portion 32L togenerate vacuum. The maximum latch suction pressure of −100 mmHg will bereached with a small amount of rebound of the tubing portion 32L and thecompression member 38 can be cycled relative to the tubing portion 32Lbetween −100 mmHg and −60 mmHg in a narrow range or band near fullcompression of the tube portion 32L. As milk flows, that narrow bandshifts at which point the tube portion 32L will be purged by fullycompressing it to drive out the contents and thereby regain morecapacity for pumping with relatively less compression of the tubeportion 32L again.

The system 10 is responsive to pressure changes within the tubing 32caused by entry of milk into the tubing 32. Referring again to FIG. 7 ,the compression elements 36 and 38 are operatively connected to a driver44, 46, respectively, for independent, but coordinated driving andretraction of the compression elements 36, 38. When electrically-powereddrivers are used, a battery 48 is electrically connected to the drivers44, 46, as well as the controller 52 and pressure sensor 54, andsupplies the power necessary to operate the drivers 44, 46 to drive thecompression and retraction of the compression elements 36, 38.

The sensor 54 is used to provide feedback to the controller 52 forcontrolling the pumping cycles to achieve and/or maintain desired vacuumlevels. Sensor 54 is preferred to be a load cell sensor providing datautilized to calculate system pressure, but could also be a pressure,flow, temperature, proximity, motion sensor or other sensor capable ofproviding information usable to monitor the safety or function of thepump mechanism of system 10. As shown, sensor 54 is a non-contact sensor54, meaning that it is not in fluid communication with the milk orvacuum space of the system 10.

As described above, the conduit or flex-tube 32 is placed in operativeconnection with a motor. An opposite side of the flex-tube 32 isequipped with the sensor 54 that takes the form of a load cell. Thepositioning of the motor is tracked and the force on the tube 34 isassessed to determine internal vacuum. By employing machine learning orsupervised learning regression techniques, the system 10 can be trainedto interpret the motor positioning and tubing strain (as well as motorspeed or pump settings), while compensating for noise and hysteresis, toarrive at a pressure/vacuum level. More specifically, a neural netsystem or any mathematical regression of the data can be incorporatedinto system firmware so that sensor input can be translated topressure/vacuum levels. In this regard, the system 10 can include orcommunicate with a non-transitory computer readable medium having storedthereon instructions executable by a computing device of the system orexternal to the system to cause the computing devices to performfunctions associated with and directed by the firmware.

To train the neural net, large amount of data is generated, both from anaccurate vacuum reading as well as strain gauge readings. All of thedata is sent to software so that post processing can be conducted. Ithas been determined that data taken during normal pump flow lends itselfbest to training the system 10. For example, data can be gathered whenflow is at 2-3 ml/min, and when system pumping is slow at each pressuretarget. This approach ensures that the motor moves along its entiretravel cycle relatively evenly and noise associated with high flow isnot introduced into calculations. Highly controlled settings are alsoused to generate data so that unbiased data is generated. Further,system accuracy can be increased when using specifically generatedneural nets for specific ranges of pressures. A special code is employedto isolate data from different pumping limits in training data, and usesonly that data to generate a neural net that is used later when pumpingto the same limits.

Turning now to FIGS. 8-10 , one embodiment of a collection or containerassembly 60 is shown. In one particular embodiment, the collection orcontainer assembly 60 can be formed from two 2.5-3.0 mil sheets ofmaterial that can be band welded or otherwise joined together along aperimeter 92 of the assembly, and can be sized to retain up to 4.5ounces, or alternatively 8 ounces of fluid. In particular, thecollection or container assembly 60 can be pre-formed to optimize ormaximize the space inside the pump system and flange. For shipping, thecollection or container assembly can be pulled closed with a vacuum tomake it flat or thin for packaging or handling. A body of the collectionor container assembly is generally bladder shaped and includes agenerally asymmetrical oval central opening 93 created by an interiorband seal. In one particular approach, the body can additionally includegussets to provide more volume. A pair of wings 94 extend into thecentral opening 93 and are provided for handling and facilitatingpositioning of the collection or container assembly 60 within a pumpsystem 10. A narrow neck portion 95 is centrally positioned and extendslongitudinally away from the central opening 93. The neck portion 95includes a tab portion 96 that provides structure for grasping andremoval, and can further include one or more cut-outs or tear-ableelements 97 provided for aiding in tearing the container 90. Furtherscoring is also contemplated to help in the tearing of the bag assembly90. Also, in alternative embodiments, the collection or containerassembly 90 can be re-sealable, re-usable, include larger or smalleropenings or include spout structure for pouring contents. A spout canalso be attached to the fitment or valve of the collection assembly tofacilitate pouring. Such a spout could further include structure whichtemporarily or permanently defeats the valve or fitment. The valve ofthe collection or container assembly can also be re-usable with a secondor subsequent collection or container assembly, and therefore isremovable from the container assembly.

Moreover, in one particular embodiment, the collection or containerassembly 90 can be made from polyethylene and can be bisphenol A free,as well as food grade material. It should be freezable without tearingand withstand approximately 0-80 degree Celsius temperatures.Additionally, tensile strength can be from 2300-2900 psi and tearstrength from 440-600 psi, with a water vapor transmission rate max ofabout 0.5 g/100 in²/24 hrs and an oxygen transfer rate of about 150cc/100 in²/24 hrs. In alternate embodiments, the material of thecollection or container assembly can be Gore-tex or Tyvek, for example.Such alternative materials can permit out-gassing. Accordingly, anon-closed or un-sealed system is also contemplated. In this specificregard, other vents or approaches to venting the system can beincorporated into one or more embodiments. Thus, self-venting of thecontainer assembly or active venting while using the pump system orafter use can be accommodated. In one approach, a pressure valve can beincorporated into the system and configured to activate after somesystem pressure is reached, and further the valve can be designed to actas a fluid barrier, only allowing air and not fluid to escape.

It is contemplated that the system is configured to pump into a sealedcollection or container assembly 60, or one that incudes an integralvalve or an otherwise airtight collection or container assembly 60, orcombinations thereof. In this specific regard, the system canalternatively or additionally be closed and never vented to theatmosphere, and/or the system suction is only reduced through the flowof milk into the system. Thus, in at least one approach, milk or fluidthat is pumped through the system is never exposed to new outside airfrom the environment once it enters the collection or containerassembly. Accordingly, the orientation of the pump system or person hasvirtually no impact on the functioning of the system (i.e., no spills).The collection or container assembly can include a rigid or flexiblesealing component, such as a ring or gasket into which the pump orcontainer valve is pushed or twisted and sealed. The collection orcontainer assembly can also include an opening or hole or structure thatis pierced such that the container assembly seals about the member thatgoes into it. Moreover, there are contemplated a range of disposable anddurable combinations of container 101 and valve fitment 102 arrangementssuch that one or both of the container bag 101 and fitment 102 aredisposable or reusable. Additionally, the container can configured to beinside or outside of the pump housing.

The fitment 102 can embody a valve such as an umbrella valve assembly103 or other type of one-way valve connected in fluid communication withthe storage container 101. The fitment can also assume a myriad ofalternative embodiments, and can additionally or alternatively be formedintegral with the container. For example, in one contemplated approach,the fitment and/or the valve can be formed as part of the containerrather than define a separate component attached to the container. Asshown in FIGS. 8-10 , however, the tail 104 of the umbrella valve 103can be employed to defeat the valve when desired such as to removegases, by turning it and engaging the tail against the valve body.Additionally, the valve includes a generally cylindrical portion havinga diameter of approximately 0.585 inches extending from a flat base 104having a width of approximately 0.875 inches. It is the flat baseportion 104 that is captured and sealed between the two sheets of bagcontainer material and includes a tail 106. The tail 106 functions toensure flow through the neck portion of the container assembly 60particularly when it is placed into the pump assembly (See FIG. 11A),and has a narrow, elongated shape that permits flow thereabout. That is,the tail 106 maintains flow through the neck even when the neck isfolded as the container assembly is attached to the breast pump body.Valve 103 prevents back flow of milk into the flex-tube 32, andfacilitates maintaining the suction (vacuum) level in the flex-tube 32.In other embodiments, other features can be provided or built into avalve to allow for depression or otherwise overcome the valve to ventair. Such approaches can involve a protrusion that is attached orassociated with the valve so that as the protrusion is pushed toward thecollection or container assembly, an edge of the valve is translated tothereby break the valve internal seal. Moreover, a nub can be attachedto valve structure and configured inside the container assembly. Tuggingon the nub through a layer of the container assembly thus results infreeing an edge of the valve and breaking the valve seal.

In at least one embodiment, the pressure at which the valve 103 opens toallow flow into the milk collection container 60 is about 25 mm Hg. Thevalve 103 can be configured and designed such that it allows fluid toflow through it when the pressure in conduit or flex tubing 32 ispositive, e.g., about 25 mm Hg, or some other predesigned “crackpressure”. The action of the compression elements cycles betweenincreasing vacuum when the compression elements move in a direction awayfrom flex-tube 32 and decreasing when the compression elements compressthe flex-tube 32, but typically should not increase the vacuum togreater than the predetermined maximum vacuum. As the compressionelements 36, 38 compress the flex-tube 32, the pressure in the system 10goes up and reaches the minimum suction level (e.g., latch suctionlevel, such as −60 mmHg, −30 mm Hg, or some other predetermined latchsuction level), at which time the compression member (pinch valve) 36seals off portion 32S thereby maintaining the minimum suction (latchsuction) against the breast. Continued compression of portion 32L bycompression member 38 continues to increase the pressure downstream ofcompression member 36, until the crack pressure is reached (e.g., 25 mmHg or some other predetermined, positive crack pressure), that opens thevalve 103. The compression elements 36, 38 continue compressingflex-tube 32, pumping fluid (milk) through the valve 103 and into thecollection container assembly 60 until the compression element 38reaches an end point in travel. The end point in travel of thecompression element 38 against portion 32L may be predetermined, or maybe calculated in real time by the controller 52 using feedback frompressure sensor 54 and feedback from the driver of the compressionelement 38, from which the controller 52 can calculate the relativeposition of the compression element 38 over the course of its travel.The compression member 36 remains closed throughout this process, as itis used to seal off the tube 32 the entire time that the compressionelement 38 is pumping milk out into the collection container assembly60. As the compression elements 36, 38 reverse direction and pull awayfrom the flex-tube 32, they start the cycle again.

As milk enters the system, the suction level decreases (pressureincreases). The feedback provided by pressure monitoring via pressuresensor 54 provides input to a feedback loop that adjusts the position ofthe compression member 38 to maintain the desired vacuum (pressure)within the conduit or flex tubing 32 by compensating for the changes inpressure that occur to changing amounts of milk in the flex tubing 32.For example, for a relatively larger amount of milk in the tubing, thiswill require a relatively shorter stroke of the compression member 38 toachieve the latch pressure. This modification can be addressed by eitherslowing the movements of the compression member 38 to achieve the sametiming cycle for pumping, or increasing the cycle frequency due to theless time taken for the shorter strokes of the compression member 38.

Use of a system 10 provided with a non-contact pressure sensor 54 wouldinclude loading the collection or container assembly 60 into the system10 (See FIGS. 11A-E). In a first step (FIG. 11B), the flange 14 isremoved from engagement with the remainder of the system 10. Attached tothe flange is the conduit or flex-tube 32. The central opening 93 isplaced over a central projection of the flange 14 and the flex-tube 32.Next, the user can pinch the wings 94 under the flex-tube 32 (FIG. 11C)followed by tucking the collection or container assembly 60 into theflange 14. The fitment 102 is placed within the collar 82 of theflex-tube 32 (See FIGS. 11A and 11E). In certain embodiments, thecollection or container assembly 60 can have useful labels, icons ornotifications. For example, milk droplet icons can be printed on thecollection or container assembly 60 in increasing size to indicate thedegree to which the container is filled, and a “this side up” messagecan be included to aid the user in properly installing collection orcontainer assembly 60. Also, the container assembly 60 includes a numberof surfaces away from storage areas where printing or handwriting can beplaced. For example, the wings 94 can be used as a writing or printingsurface as can the pull tab 96. Similar labeling or messages (forexample, a “thanks mom” message) can be included on the collar 84 orother portion of the flex-tube 32 to aid in properly orienting theflex-tube with respect to the flange 14. It is to be recognized that thecollection or container assembly can be placed in alternative locationsas well. For example, the collection or container assembly can beconfigured around the nipple of the breast. In this regard, thecontainer assembly itself can form the desired flange or breastcontacting structure into its core construction. In one specificapproach, the container assembly can also include more surface areafacing the breast superior to the nipple than inferior thereto.

It is contemplated that the door assembly 90 be employed to both providea continuous contour of the flange 14 for engaging a user's breast aswell as to support the engagement of the collection or containerassembly 60 with the system 10. Thus, the door assembly 90 can beconfigured to pivot with respect to the flange 14, and employed to closethe system 10 as it is snapped over and closes the pump. With thisapproach, the fitment 102 and container bag 101 are securely sandwichedbetween the collar 82 of the conduit or flex-tube 32 and the doorassembly 90, with a cylindrical portion of the fitment 102 receivedwithin the collar 82. The collar 82 can also provide rigidity to theflex-tube 32 so that it can be loaded into the flange 14 as well as toprovide an annular back-up when the fitment 102 is inserted. Ribs oro-rings can be provided on an interior surface of the flex-tube tofacilitate sealing with the fitment 102 and can have a radius ofapproximately 0.64 mm. In one embodiment, the interior diameter of theflex-tube between the ribs can be approximately 14.6+/−0.17 mm, whilethe outer diameter of the fitment 102 can be on the order of 14.8+/−0.17mm so that an interference fit results, with a force of around 1-2.5lbs.

As best seen in FIGS. 12 and 13 , the door assembly 90 further includesa pair of spaced and curved guiding arms 105. The arms 105 are contouredto guide the door assembly 90 as it closes over and about a pair ofcurved railings 107. In this way, as the door assembly 90 rotates towardthe conduit or flex-tube 32, a latch 109 of the door assembly 90 firstclears and then seats beyond and behind the railings 107 to therebyprovide a robust engagement with the flange 14 and bag assembly 60 whenit is loaded in the system 10 (See also FIG. 11 ). As best seen in FIGS.8 and 9 , the fitment 102 includes a scalloped portion 110 whichfunctions to facilitate this robust engagement and provide relief ofundue stresses being placed upon the bag assembly when securely mountedwithin the system by resisting kinking. The door assembly can furtherinclude one or more ribs 111 (FIG. 15 ) that engage and provide directedsupport to the fitment 102.

Turning now to FIGS. 16-18 , there is shown an approach to protectingagainst pinching by moveable parts of the pump system 10. One or moremagnets 118 can be attached to the flange 14. A corresponding sensor 119(such as a hall effect sensor) can be attached to the flex circuit 16that is mounted to a mounting bracket 120 (See also FIG. 3 ). The system10 can be configured to only permit the motor to be activated when thesensor detects the magnet 118. In this way, the pumping action of thesystem and in particular, the compression members will not move untilthe flange 14 is attached properly to the housing 12, consequentlyavoiding any pinching or engagement of such components with the user. Inother approaches, mechanical or electronic switches or RFID technology,or optical sensor or sonar technology can be incorporated into thesystem to provide the desired safeguards, such that the system will notoperate unless all components of the system (i.e., flange, tubing andstorage) are fully connected.

In another approach, the system 10 can include firmware that operates totrack system pressure on a load cell. Here, the motor paddle can bearranged and controlled by firmware so that it is moved outwardly 0.5 mmor some defined distance and pressure on the load cell is observed tosee if the conduit or flex-tube is properly installed. Where thepressure observed is not as expected, such as if there is no pressure,the motors will not be permitted to move inward for pumping. Using asimilar technique, proper collection container install can be tested.After the motor extends, the pinch can seal the flex-tube. Once themotor comes back out, a vacuum will only be measured if the bag isproperly installed, sealing air from filling the tube on the containerside.

It has also been recognized that fluid ingress protection may benecessary for the pump system. Thus, it is contemplated that variousgaskets can be configured within the structure of the system. Oneparticular location for a gasket is the interface between the chassisand housing, and accordingly, a specially designed gasket is configuredabout a perimeter of the chassis along a section which is intended toengage the housing. In this regard, a 0.3 mm interference fit iscontemplated between the gasket and the housing. Also, gaskets can beconfigured about moving receiver structure such as that of the load celland motor to help prevent fluid ingress.

Once the flange or skin contact member 14 is placed onto the mainbody/pump housing 34 then pump power can be engaged. Referring now toFIGS. 19-20 , there is shown an enlarged view of the user interfacepanel which as stated above includes a button membrane 72 and a buttonmembrane housing 73 each supported on the housing 12 and placed inengagement with the flex-circuit that provides the user with systemcontrol. Here, the membrane 72 acts as a light pipe. Light emission,intensity and button deflection forces are configured for convenient andeffective interaction by the user. Accordingly, pressing the powerbutton 130 functions to start the pump system 10 through its interactionwith a switch 131 configured on the flex circuit 16 (See FIG. 18 ). Itis noted that other switches 132 can further be provided on the flexcircuit which line up with other contemplated system control buttonsincluded on the flex membrane 72. Should the system 10 require externalpower or the battery charged, access to supporting electronics is gainedthrough the cover jack 65 that is configured in the shell 12 (FIG. 21 ).

As the pump system 10 goes through a power up routine, the controller 52reads force on the load cell when a load cell is used as the pressuresensor 54. This is the load measured by the load cell, before the skincontact member 14 has been applied to the breast, so in one approach itis a state in which the pressure in the conduit or flex-tube 32 isatmospheric pressure. The controller 52 then calibrates the system suchthat the preload force or position or measured load or strain equates toatmospheric pressure. Based upon a neural network or computer learning,load or strain detected at the flex-tube 32 can be converted to pressurereadings in the system 10 during operation of the breast pump system 10upon attachment to the breast.

The system 10 can calculate the volume of milk pumped into system oralternatively the volume collected in the milk collection containerassembly 60. By knowing the dimensions of the conduit or flex tubing 32downstream of the compression member 36 when compression member 36 hassealed off tubing portion 32S, the overall volume capacity of the system10 downstream of compression member 36 can be calculated. With referenceagain to FIG. 7 , tracking of the position of the compression member 38relative to the tube 32 (such as by knowing the driver 46 position atall times, for example), dictates the volume change in the tubing 32. Asthe pumping process is carried out, pumping/purging of milk into themilk collection container occurs when the compression member 36 hasclosed off the small tube portion 32S at the location of compression.When the compression member 36 has closed off tube portion 32S, thechange in position of compression member 38 that occurs to carry out thepurge of milk from the flex tubing 32 and into the milk collectioncontainer 60 is used to calculate the change in volume of the tubing 32downstream of the compression member 36, which equates with volume ofmilk that is pushed into the milk collection container 60 bag.

In particular, under one algorithm, as flow enters the system 10, it isrecognized that the motor must move further and further out to generatelatch vacuum. Tracking this movement and the rate of position change ofcompression or paddle members as they generate latch vacuum is one wayof measuring flow. The slope of a line that is associated with trackingpaddle locations for latch vacuum, for example, is directly proportionalto flow. After a tuning step to correlate this relationship, calculatingflow from the slope of this line can be readily accomplished.

Using another approach, the number of purges can be tracked when thesystem is full for the purpose of measuring flow. As stated, it can bedetermined when the system 10 is purging fluid versus purging air sincethe forces are much higher for purging fluid than purging air. Thus,counting the number of purges that contain fluid, and knowing the volumethat is purged for each purge leads to a calculation of flow withoutrequiring significant system tuning or calibration, and avoidingconfusing a slow air leak with flow. Leaks can also be detected byemploying an algorithm involving closing the pinch compression member,followed by closing the pump compression or paddle member, and thenpulling the pump compression member outwardly to create a vacuum. Bythen holding the pump compression member in this position and verifyingthe vacuum is maintained, it can be determined if there is a leak in thesystem 10.

In addition to calculating the volume of milk purged with each purgecycle, the system (via controller 52) can sum the volumes from all purgecycles to calculate the total volume entering the pump or alternativelypushed into the milk collection container 60 during a milk extractionsession. This volume can be stored with a unique identifier provided tothe milk container so that the system 10 keeps a record of how much milkis stored in each milk collection container 60. This information canalso be time stamped so that the user will know the time and date thatmilk was collected, regarding each milk collection container. Additionalstatistics can be calculated, including, but not limited to: averagevolume per extraction session, total volume extracted for any given day,average milk extraction volume per day, etc. Any and all of this datacan be exported to an external computer, either manually, or it may beautomatically uploaded to the computer when the computer is within rangeof the system 10 for wireless communication, or when the computer isconnected to the system by wire. Further optionally, any or all of thisdata can be either manually or automatically uploaded to a cloud serviceover the Internet, either wirelessly or by wire.

In one preferred approach, volume that is extracted from a breast iscalculated by building a map of motor location versus volume at aspecific vacuum such as −60 mmHg, since the same is a reliable andpredictably repeatable vacuum level. To build this relationship, variousknown rates of flow are created within the breast pump, and motorlocations at −60 mmHg are identified and stored as data. A script isthen employed to extract such data to build this motor versus volumescale. In particular, as shown in FIG. 7B line M represents motorlocation versus time and line V represents vacuum versus time. This datais entered into the script to arrive at a graph of volume over time. Thescript looks at indications of times of purge and tosses out data excepta first clean start and finish of a cycle between purges represented bythe spike S in FIG. 7B. An adjustment is made for time such thatrelative time for the data selected replaces absolute time. The scriptalso functions to filter out data that is too far above or below −60mmHg. Relative time is then turned into volume in that motor locationbecomes volume difference over a relative time span. This data can beplotted as a line (See FIG. 7C), the area under which represents volume.The output of the script represented as a line having a particular slopeis expressed as numbers and as a 6^(th) order equation representing theline. This 6^(th) order equation is incorporated into the code base andis employed during real time to convert measured motor locations duringpumping at approximately −60 mmHg into volume. Calculus is thusultimately employed to arrive at this volume calculation in that motorlocation versus volume is integrated between two points to determinedifferences in volume. At certain levels of flow there can be anadjustment factor multiplied by volumes calculated when comparingcalculated flows with real world experimentation, in particular toaccommodate for flows shortly before and after purging, for example, orfor flow above 15 ml/minute. Additionally based upon empiricalobservations, when the pump is filled with air, a mathematical constantis incorporated into the volume calculations. A determination that thepump is filled with air can be made by observing how hard the motor hasto work, and knowing that the motor has to work harder to create vacuumchanges when the system is filled with air as opposed to when the systemis fluid filled.

When calculating milk volume pumped from the system 10, as stated, thereis a need to distinguish between any air pumped by the system versusmilk pumped from the system, as well as pumping mixtures of milk andair. When initiating a milk pumping/extraction session, there is air inthe tubing 32 this initial volume of air needs to be pumped into themilk collection container 60 to prime the pumping system 10. Again,distinction between pumping air versus pumping milk can be recognized bycorrelating pressure changes with the amount of movement of compressionmember 38 needed to establish the pressure changes. For example, whenair is in the tubing, a greater change in position, or more overalltravel of the compression member 38 is needed to establish the samepressure change than that needed when the tubing 32 is filled with milk.Thus, relatively more motion of the compression member with relativelyless pressure change indicates air in the tubing 32. This difference inpressure may also be detected when the compression member 36 is open(i.e., not closing off tube portion 32S) and compression member 38 isretracting and this increasing the vacuum pressure.

When a user has completed the pumping phase of extracting milk from abreast, it is useful and efficient to purge as much milk that remains inthe tubing 32 from the tubing 32 and into the milk collection container60. Ending of the extraction phase can be performed upon elapse of apredetermined extraction phase time, calculation of a predeterminedamount of milk having been pumped, manual cessation of the extractionphase by the operator, or some other predetermined value having beenachieved after performing the extraction. The direction of the pumpingstroke of compression member 38 is reversed and the compression member38 is run in the reverse direction to decrease suction within the tubing32 and optionally create a small positive pressure within the tubing 32to facilitate removal of the system 10 from the breast. Alternatively,the suction may be decreased to a level where a slight suction remainsso that the user still pulls the system 10 of the breast to detach it.Preferably the vacuum is reduced to 0 mmHg, or a slightly positivepressure to automatically detach the system 10 from the breast. The endpressure value where the pressure reduction by reverse pumping is ceasedcan be in the range of about −20 mmHG (weak vacuum) to a positive 50mmHg (e.g., the crack pressure of the valve to the container). Thecompression member 36 does not close off the tubing portion 32S duringthis process, rather, tubing portion 32S remains open. Initiation ofthis reverse pumping may occur automatically or, alternatively, may beinitiated by the user. This process continues until the seal of thesystem 10 to the breast is broken, which is detected by the controllervia sensor 54. Once exposure of the tubing 32 to atmospheric pressure isdetected, the stroke direction of pumping is again reversed therebypumping the milk in tubing 32 under positive pressure and driving themilk from the tubing 32 into the container 60. If by chance, the system10 accidentally or otherwise becomes resealed to the breast during purgepumping, the system 10 can automatically shut down as it senses vacuumpressure being regenerated in the vicinity of the flange or breast/skincontact member 14.

The system 10 can be configured to distinguish whether it has beenattached to the left breast or the right breast of the user. This can beuseful for tracking milk volume output per breast, per session, totaldaily volume per breast, etc. When using two of the pump systems, thetracking of data for each breast can still be maintained accurately,even when one of the pump systems 10 is attached to the left breastduring a current pumping session after having been attached to the rightbreast during a previous pumping session. In one embodiment, the pumpingsystems 10 can establish current location (i.e., left or right breast)by receiving a signal from the other pumping system having been attachedto the other breast. This established relative left-right locations ofthe two pumping systems 10, so that each system 10 can accurately recordas to whether milk is being extracted from the right breast or leftbreast. This identification is automatic, without any user inputrequired and it also relieves the burden on the user to otherwise keeptrack of which pump system 10 is placed on each breast and to maintainthis order with each successive pumping session. Left and right pumplabeling is also contemplated such as by placing markings on the systemhousing or cover jack, for example, near the power connector.

Various approaches to assessing milk volume can be included in the pumpsystem. Certain approaches are describe in co-pending InternationalApplication No. PCT/US15/50340, the entirety of contents of which areincorporated herein by reference. A further approach to assessingexpressed milk volume involves placing one or more disposable datacollection devices on the mom or child. One specific approach involvescreating a boundary on the skin of a breast and employing a fiducial toconveniently measure the change in size of the boundary. This change insize is then correlated to milk production to arrive at a volume of milkexpressed or pumped. A crib or bassinet can also include sensors andcommunication hardware that communicate with the pump system so bothassess and management milk consumption and needs, and baby health.

The system 10 can calculate the pressure during operation in any of themanners described above. The suction (pressure) level can be varied asdesired, and by continuously or repeatedly measuring/calculatingpressure, the feedback provided by sensor(s) 54 to controller 52provides a control loop that can be used to adjust the compressionmember 38 position and/or speed to vary the suction pressure to a leveldesired, or maintain a desired suction pressure in real time. Thus,controller 52 can control the positions and speeds of compressionmembers 36, 38 to achieve any vacuum pressure pumping profile desired,and provide automatic, real time adjustments to maintain a desiredvacuum pressure within the system. Also contemplated is responding inreal time to maintain flow. This can accomplished independent or inconjunction with monitoring and regulating pressure in real time.

The controller 52 tracks the position of the compression member 38relative to the tubing 32L, such as by keeping track of the driver 46position or shaft position (interconnecting linkage between driver 46and compression member 38), and calculates (or looks up) pressure basedupon data received from sensor 54. The system controller or firmware isprogrammed with or retains information relating values detected bysystem sensors with driver positions and speed and system pressure.Thus, changes in position and/or speed of the compression member 38 bycontroller 52 can be controlled by resulting changes in pressurecalculated or looked up, relative to the pressure sought to be achieved.As stated above, by using machine learning or supervised learningregression techniques, the system 10 can be trained to interpret themotor positioning and tubing strain (as well as motor speed or pumpsettings), while compensating for noise and hysteresis, to arrive at apressure/vacuum level. More specifically, a neural net system or othermathematical regression can be incorporated into system firmware so thatsensor input can be translated to pressure/vacuum levels. Controller 52can thus control compression member 36 in a similar manner, but controlof member 36 is more focused on position control, as the compressionmember 36 needs to fully close off tube portion 32S when maintaininglatch suction against the breast/nipple. However, the closing off istimed and performed at the determined latch pressure, which is knownfrom the data received from sensor 54.

Turning now to FIGS. 22-28 , various aspects of remote control and datacollection approaches are presented. In at least one contemplatedembodiment, the system 10 can be configured to communicate with aserver, a remote computer, smartphone or other device such as throughsignal, such as by Wi-Fi, BLUETOOTH, BLUETOOTH Low Energy (BTLE), RFID,NFC or the like. In particular, one or more chips can be incorporatedinto the controller of the pumping system 10 (by hard wire and/orwirelessly, preferably wirelessly) and configured to be in communicationwith an external computer. The controller and/or external computercommunicates with the sensor(s)/chip(s) which indicate(s) when thesystem is in use, and can track usage. By tracking the times of useand/or number of uses, or even pump cycle counts, for example, thecontroller, or external computer can alert the user when it is time tochange components or to report on usage aspects. In this way,information such as the tracking of extraction date and time, volumeextracted, etc. can be recorded and stored with regard to each milkcollection container used with the system 10 to extract milk. Thus, thesystem 10 can register individual milk collection containers, so thatthe user can readily identify when milk in each container was collected,the volume in each container, etc. The breast pump system can record thevolume of milk in any given container during a pumping session. The datarecorded can be sent to an external computer and/or over the Internet,either automatically or manually. Thus, user data and trends can becollected, stored and analyzed as they relate to volume (from eachbreast and in total), as can be the number of sessions on severaldimensions (per day, week or month). Data and analytics can thus beprovided to a user concerning pumping sessions.

In one particular approach, at least the session start time, the sessionend time and total volume of milk extracted from the breast can bestored and tracked. Sessions can be defined as the commencement of latchand can continue up to and through pauses of up to 5 minutes, forexample. Thus, a pause of over 5 minutes can be defined as the end ofthe previous session. A language protocol is generated so that there isa two-way communication between an external device or program and thebreast pump. That is, both the pump and external device can create andunderstand and are responsive to specific messages. Further, live dataand historical data can be treated differently, and their data streamsmaintained separately. Live updates are generated and stored at the pumpand are available by the external device to retrieve (for example, up ordown button activations or volume updates). Accordingly, such live datacan be reflected on and update the screen of the external device.Historical data is stored inside the pump in a stream and the pump cancommunicate with this stream to extract or act upon the same. Aninternal pump memory such as a disc within a chip or other internalflash stores, communicates with the pump so that session data is writtento an internal history log. At the end of a session, for example, thepump will write the session data to its internal history log and theexternal device will ask if there is any data and if the pump indicatesthat there is, then the external device will download this historicaldata to update its non-live view screens. The external device can alsomake this same query after an extended time and then download multiplesession data, and the query also can be made during a session. In oneparticular embodiment, as much as 600 sessions of data can be stored.

In one or more embodiments, the system can additionally or furtherinclude structure configured to accomplish or functionality operating asan Active Pause Mode, that allows the system to maintain latch vacuum,while remaining (especially under no/low flow) virtually silent. Such asystem stays much quieter than pump mode, but ensures the system doesnot fall off the breast. It can be employed by the user mom when sheneeds to interact with others and does not want them to hear the pump,or for some other reason where she might not be ready to remove thedevice but does not want active pumping either.

The remote user interface 140 on an external device can take a myriad offorms. The pump system can also be personalized such as by giving one ormore pumps a name (FIG. 22 ). A user profile can be created for a childand linked to a child's birthdate (FIG. 23 ). Other details such astracking the child's age when use of the system commenced can begathered so that analytics pertinent to the child's age can begenerated. In this way, pump performance can be tracked to the growth ofthe child. Reminders can be entered into the system (FIG. 24 ) so thatthe user can focus on matters other than breast pumping. Notificationscan be keyed to time or volume of milk pumped while both of suchcriteria as well as battery life can be tracked and reflected on theremote computer (FIGS. 25-28 ). Easily understood and convenientgraphics are contemplated for expressing status such as curvedhemispherical strips 150 reflecting volumes pumped for each pump system,the same information also being shown in numerical form 152. Timingcountdowns as well as information from one or more previous sessions canalso be graphically displayed for effectively communicating with a user.The ability to remotely begin a new session can also be made availableto the user.

The remote user interface 140, whether provided as an App, on a cellphone, computer or other computing device, can also include specificuser control functionality, and various related easy to understanddisplays (See FIGS. 29-41 ). As shown in FIG. 29 , in one or moreapproaches, the amount of milk pumped is tracked by day, and an optionis provided to the user to set a session tracker by day. The amountspumped are also tracked by breast. A user can set one or more of timeand volume of pumping by breast for one or more pumping sessions. Forexample (FIGS. 30-31 ), a volume target can be set by the user byvarious increments such as 0.1 ounces. This setting can be set andsaved, or canceled. As shown in FIG. 32 , the user can then controlwhether to pump with one or both breasts, and then the system startstracking pumped volume (see FIG. 33 ). As pumping progresses (FIG. 34 ),easily readable curved bars reflect the amount of volume being pumped byeach breast, the bars becoming thicker as more volume is pumped. Theuser can adjust suction levels for one or both of the pumps attached toa breast (FIGS. 35-36 ) to coordinate pumping or to otherwise pump asdesired. After reflecting the changes in suction level, the user canreturn the system to tracking volumes pumped by breast (FIG. 37 ), andan indication of volume remaining to be pumped is also provided. Oncethe pumping target is met, such as a target volume (FIG. 38 ), the userinterface will indicate that the session has been completed. Thereafter,an updated set tracker is presented, with an ability to set furtherpumping schedules (FIG. 39 ). The user can then select an option todepict a summary of pumping or a history of pumping (see FIGS. 40 and 41). The data provided by the user interface can include bar graphs andnumerical data showing pumping by day and by breast and session timesand number of session. Additionally, circles can be sized to representrelative amounts of pumping by date, and color coated by breast.

The pump system can also include a power management system thatfunctions to save power. In one aspect, the pump system can becharacterized as having multiple modules or threads, each runningseparate programs. Each thread, such as fifteen to twenty differentthreads, is designed to operate in a manner to save power. That is, eachthread is controlled so that it seeks and finds its own maximum, minimumrequired power mode.

The pump system can be further configured so that the power managementsystem includes a power hierarchy including various different levels atwhich the threads seek to achieve maximum, minimum required power. Inone approach, the levels can include one or more of hibernate, standby,standby with LED's and active. Hibernate can be characterized as a deepsleep state and standby can be defined at a level where the system isrunning the computer chip and running calculations but not externalcomponents are being run. Standby with LED's can mean just that theLED's are engaged and active can mean that external components like themotors and sensors are working. The power system can thus function sothat a query is sent to each thread asking for the thread's currentstate and it's minimum required mode. The power system then cyclesthrough each thread and sets the power level at the maximum, minimumpower level required so that each thread can properly operate.

In still yet further embodiments and approaches, the pump system canalternatively or additionally include built-in or computer or App basedfunctionality to de-stress the user's life, empower the user to bettertake care of the nursing baby's health, maximize the user's mobility andfreedom, and support all that is involved in becoming or being a parent.In these regards, pump system structure and functionality can includeone or more of keying on pain points, physical conditions, sleep, painrelief, and post partum issues, tracking sleep, sensing and trackingbaby vitals and movement, focusing on connected health with the mom asthe caregiver, and/or providing education, guidance or instruction onmovement and ways to carry a baby, fertility, post baby needs, health ofthe mom, ultrasound and fertility. The pump system can additionallyinclude App integration with smart bottles, smart scales or the like tofacilitate managing overall baby health and nutrition. App updates canadditionally be provided about stimulation and letdown, and timing ofpumping based upon such information, such as suggestion to beginpumping. System structure and functionality can also involve updatingpumping profiles based upon baby age and needs, developing pumpingfunctionality which enhance milk production, enhance efficiency orcomfort or better mimic baby. Data can be stored in the cloud foranalysis, and additional functionality can be provided to modify speedsand alternated between and among customized modes and profiles.Additional or a myriad of sizes of flanges and bag or containerassemblies can be provided to the user as can nighttime pumpfunctionality or programming including automated sessions with startsand stops.

Inventory management is further functionality that is provided as partof the structure of the pump system. In connection with the same,container assemblies can include structure that is scannable or whichotherwise communicates with the inventory management system (eg., viabar codes, RFID chips). Further, operative communication structure canbe provided so that the user can transmit data with and between ababy-center platform that stores data thus facilitating an avenue forthe effective management of the baby's nutrition, and links can be madeto automatically communicate with milk banks and donation centers.Additionally, a caregiver data share system can be included within thefunctionality and structure of the pump system. Texting is added toother forms and avenues for communicating such important and usefulinformation.

While the present disclosure has been described with reference to thespecific embodiments thereof, it should be understood by those skilledin the art that various changes may be made and equivalents may besubstituted without departing from the true spirit and scope of thedisclosure. In addition, many modifications may be made to adapt aparticular situation, material, composition of matter, process, processstep or steps, to the objective, spirit and scope of the presentdisclosure. All such modifications are intended to be within the scopeof the present disclosure.

That which is claimed is:
 1. A wearable, portable self-powered breastpump system sized and shaped to fit within a bra and configured forpumping milk from a breast, comprising: a breast adapter configured tocontact and form a seal with the breast; a nipple receiving cavityconnected to the breast adapter; a pumping mechanism including a motorfixed with respect to the nipple receiving cavity and configured to pumpmilk from the breast; and a milk collection container, the milkcollection container configured to surround the nipple receiving cavity.2. The system of claim 1, wherein the breast pump system automaticallysenses letdown.
 3. The system of claim 1, wherein the breast pump systemswitches from letdown to extraction upon sensing a small volume of fluidhas been collected.
 4. The system of claim 1, further comprising anon-contact pressure sensor adjacent the nipple receiving cavity.
 5. Thesystem of claim 4, wherein the non-contact pressure sensor is used forcontrolling pumping cycles of the breast pump.
 6. The system of claim 1,wherein the breast pump includes a letdown phase and an extractionphase.
 7. The system of claim 6, wherein the extraction phase is for apredetermined amount of time.
 8. The system of claim 1, wherein themotor is fixed above the nipple receiving cavity.
 9. The system of claim1, comprising two breast pumps, wherein each breast pump has left andright pump markings.